Calibration Control. Antenna have no influence / SOLVED
Selecting the antenna in the Calibration Control doesn't seem to affect the measurement curve in the Spectrum View.
Setting Cable and Attenuator selection right away adjust the shown curve.
No change happens by Antenna selection.
A pseudo antenna was defined (-20dB from 1Mhz to 1Ghz). No change either.
I think you use the wrong measurement unit: Please note that an antenna can only effect field strength measurements (V/m, dBµV/m etc.), it would make no sense with a standard dBm view so there will be no change with this unit.