3D vs XYZ measure with LCS software and NF5030
I'm using the NF-5030 linked to a PC with USB cable. I run the LCS Analyzer software.
The instrument is near a crt monitor.
Start = 50 KHz
Magnetic Axis= X
The LCS screen shows a signal centered at 110KHz. Its value is 60 nTesla.
If I change the measure to 3D (instead of single X axis), the NF5030 seems not work. The sweep in LCS starts to 50 khz, then goes to about 55 KHz and do not go up. It returns to 50KHz and the cicle begin agains. If I change the sampletime to 200msec, it works again.
Does the 3D measure need a shorter sampletime?